An artificial sphincter consists of three parts:
To treat urinary incontinence, the cuff is placed around the bladder neck so that when it is inflated, the urethra will close tightly. The pressure regulating balloon will be placed under the tissues of the lower abdomen. The balloon is filled with a liquid (occasionally, an iodine-based solution will be used so that it will be visualized when x-ray procedures are performed). The control pump mechanism is placed in the labia for women, and in the scrotum for men. To use the sphincter, the person will compress (squeeze) the pump to divert fluid from the urethral cuff to the balloon. This action will allow the sphincter to relax so that the person can urinate. The cuff will then re-inflate on its own in 3 to 5 minutes .
Updated by: Linda J. Vorvick, MD, Medical Director and Director of Didactic Curriculum, MEDEX Northwest Division of Physician Assistant Studies, Department of Family Medicine, UW Medicine, School of Medicine, University of Washington; and Scott Miller, MD, Urologist in private practice in Atlanta, Georgia. Also reviewed by David Zieve, MD, MHA, Medical Director, A.D.A.M. Health Solutions, Ebix, Inc.
Related MedlinePlus Page
The information provided herein should not be used during any medical emergency or for the diagnosis or treatment of any medical condition. A licensed physician should be consulted for diagnosis and treatment of any and all medical conditions. Call 911 for all medical emergencies. Links to other sites are provided for information only -- they do not constitute endorsements of those other sites. Copyright 1997-2014, A.D.A.M., Inc. Duplication for commercial use must be authorized in writing by ADAM Health Solutions.